Payment & Shipping Terms:
|Test Speed:||0～30 Times/min||Times Setting:||0 ~ 999,999|
|Control Mode:||PLC And LCD Touch Screen|
furniture impact tester,
Customized Chair Drop Impact Tester,
impact furniture testing equipment
BIFMA X5.1 Customized Office Chair Drop Impact Test Machine
Test rig for chair drop durability is used to evaluate the ability of the chair to withstand heavy and abusive impact forces on the seat. It can also to evaluate the ability of chairs to withstand fatigue stresses and wear caused by downward vertical force(s) on the seat.
Machine Dimensions: (W x D x H) 1600*1500*2400 mm
Power supply source: single-phase, AC220Volt, 50Hz
Gross Weight: 820kg
|Adjustable Drop Height||10mm to 180mm|
|Largest chair dimensions width||1000mm|
|Test speed||0～30 times/min|
|Times setting||0 ~ 999,999|
|Control mode||PLC and LCD touch screen|
total test bag weight 136kg
(can be divided to 136kg 102kg, 57kg)
BIFMA X5.1 2017 section 7 Drop Test-Dynamic & section 10 Seating Durability Tests-Cyclic.
Test rig for chair drop test according to BIFMA X5.1 2017 Clause 7 & Clause 10.3.
To evaluate Chair to withstand fatigue stresses and wear caused by free fall vertical forces on the seat.
BIFMA X5.1 2017 Clause 7
==Functional Load Test
a) A test bag weighing 102 kg (225 lb.) shall be raised 152 mm (6 in.) above the uncompressed seat and released one time.
b) Remove the bag.
c) For chairs with seat height adjustment features, set height to its lowest position and repeat a) and b).
==Proof Load Test
Similar with Functional Load Test, but the force is 136 kg (300 lb).
BIFMA X5.1 2017 Clause 10.3
Impact load: 57kg(125lbf)
Impact height: 36mm(1.4in)
The chair shall be tested to 100,000 cycles.
A) Quality assurance items
The guarantee period with FOC within one year (not including nonexpendable & transport and travel fees)
B) The main quality warranty maintenance certificate
If there is any dispute please refer to our guarantee letter, so:
① please keep the guarantee letter, if you lost it, please connect with us in a month.
② If the guarantee letter has been altered or it has no our stamp, it is useless.
C) The following conditions need to be paid reasonably even in the assurance period:
① Natural reasons
② Operating mistakes
③ Voltage is not fit for our operation instruction
④ Repack it without our guides
⑤ Damaged for borrowing to others
⑥ Damaged for authorized machine modification
⑦ Damaged for authorized calibration
⑧ Authorized transshipment mistake
⑨ Serve for long distance area
Contact Person: salesmanager