Payment & Shipping Terms:
|Test Types:||Light Only, Dark Only, Spray Only||Total Exposure Area:||1370cm2|
|Test Positions:||11 (ISO) Or 5 (AATCC)||Irradiance Monitoring System:||Radiometer|
|Irradiance 420nm Range:||0.80~2.01W/m2||Rack Rotating Speed:||5r/min|
textile testing instruments,
textile industry equipment
Textile Testing Machine Electronic Xenon Lamp Air Cooled Light Fastness Testing Machine for Textile
This machine designed to meet the requirements of ISO 105 B02 and AATCC Test Method 16 standards. It has a rotating specimen rack and provides precise control of critical test parameters including spectrum, irradiance, relative humidity, chamber temperature and black standard temperature.
|Light Source||2500W air-cooled xenon arc lamp (recommend replacing lamps every 1500 hours)|
|Test types||Light only, Dark only, Spray only|
|Total Exposure Area||1370cm²|
|Test Positions||11 (ISO) or 5 (AATCC)|
|Exposure Area of each Test Position||ISO 65.25cm² (145 x 45) mm
AATCC 108.75cm² (145 x 75) mm
|Irradiance Monitoring System||Radiometer|
|Irradiance 420nm Range||0.80~2.01W/m²|
|Rack rotating speed||5r/min|
|Test Chamber Humidity Range||Light Cycle 10%~70%RH,Dark Cycle 30%~95%RH|
|Test Chamber Temperature Range||25ºC~50ºC|
|Black Standard Temperature Range||40ºC~85ºC±1ºC|
|Black Panel Temperature Range(optional)||40ºC~80ºC±2ºC|
|Water spray||specimen surface|
|Water Reservoir Capacity||15 liters|
|Water Consumption (ISO 105-BO2 Normal)||0.9 l/h (max.)|
|Color Touch Screen||Supplied as standard|
|Chamber Viewing Window||Supplied as standard|
To determine the resistance of the color of textiles to fade when exposed to artificial light (xenon arc) which is representative of natural daylight.
ISO 105-B02 Textiles -- Tests for color fastness -- Part B02: Color fastness to artificial light: Xenon arc fading lamp test
ISO 105-B04 Textiles -- Tests for color fastness -- Part B04: Color fastness to artificial weathering: Xenon arc fading lamp test
ISO 105-B07 Textiles -- Tests for color fastness -- Part B07: Color fastness to light of textiles wetted with artificial perspiration
AATCC TM 16 - Option 3 Colorfastness to Light: Xenon-Arc
M&S C9 Colorfastness to Light
M&S C9A Colorfastness to wet light in the presence of alkaline or other chemical residues
JIS L 0843 Test method for color fastness to xenon arc lamp light
JIS L 0888 Test method for color fastness to light and perspiration Method B
IS 15203 Textiles - Method of Test for Determination of Color Fastness of Textile Materials to Artificial Light at High Temperatures - Xenon Arc Fading Lamp Test
1. Full Color 10.4" Touch Screen Control Panel Display of
Direct Setting and Control of Irradiance
Direct Setting and Control of BPT/BST
Direct Setting and Control of Relative Humidity
Direct Setting and Control of Specimen and Chamber Air Temperature
8 Factory Pre-programmed Test Methods
Space for 4 Custom Programs
2. Graphic display of the progression of all test parameters.
3. High performance, long lasting, air-cooled xenon lamp.
4. Solar-powered radiometer, incorporating a black standard thermometer.
5. Works in "controlled irradiance" or "blue wool" mode.
6. Eleven double-sided sample holders offer a massive exposure area of up to 1370cm².
7. Refined software which perfectly reflects the needs of the operator.
8. High efficiency power supply ensures running costs remain low.
9. Complies with international standards for textiles, leather and other materials.
10. Compact, stylish and ergonomic design.
Accessories supplied with the instrument
|1.Main Unit||01 No.|
|2.Test Kit 7.3cm² (Ø 30.5mm) and 50cm² (Ø 79.8mm)||01 No.(each)|
|3.Synthetic Rubber Diaphragms(Maufacturer||James Heal)||05 Nos.|
|4.Diaphragm Unit Tightening wrench||01 No.|
|5.Air Pipe (To connect with compressor)||01 No.|
|6.Glycerol (500mL)||01 No.|
|7.USB software||01 No.|
|8.Power Cord||01 No.|
|9.This User's Manual||01 No.|
|10. Inspection & Conformance Certificate||01 No.|
Contact Person: salesmanager